13 BMD Tomography (CMT) Microtomography : Typically monochromatic beam 10-65 keV, sometimes white beam to 100 keV. Maximum beam size is 50mm (1mrad). Contact : Mark Rivers, email@example.com.
13 BMD Diamond Anvil Cell (DAC) Diamond anvil cell diffraction and imaging : Typically run at 37 keV, but sometimes up to 70 keV. Maximum horizontal beam size of 5mm (0.1 mrad). Contact : Vitali Prakapenka, firstname.lastname@example.org
13 BMD Multi-anvil Press (LVP) Multi-anvil press diffraction and imaging : Typically run white beam energy dispersive diffraction with signal needed to 100 keV. Monochromatic diffraction and imaging at 65 keV. Maximum beam size 10 mm. Contact : Yanbin Wang, email@example.com
13 BMD X-ray Absorption Fine Structure (XAFS) Spectroscopy : Energy range 5 - 45 keV, beam size to 10mm. Contact : Matt Newville, firstname.lastname@example.org
The bending magnet fan is divided into two parts. An outboard fan of 2.5 mrad feeds the 13-BM-D station, and an inboard fan of 1.5 mrad feeds the 13-BM-C station. These fans are separated by 1 mrad. The extreme inside and outside 0.5 mrad on each edge of the 6 mrad fan is not used.
The 13-BM-D beamline has a non-focusing Si <111> double crystal monochromator that covers the energy range from ~5 to 70 keV. There is a 1 m long vertical focusing mirror in the 13-BM-B optics station. This mirror is used for some experiments and removed for others. There are small K/B mirrors that are placed on the experimental table in 13 BM-D.